## Description

## Precision source and measure for thermocouple, mA, mV, and V devices

412400 Multifunction Process Calibrator sources and measures Thermocouples (J, K, and T), mA, mV, and V devices. The 24V Loop Supply drives current loads up to 1000 ohms. Memory for 5 user settable output values of each type for fast calibrations. Single/continuous step function for Voltage and Current. Comes complete with six AA batteries and calibration cable with spade lug terminals.

Specifications | Range | Accuracy (of reading) |
---|---|---|

DC Source | ||

Current | 0 to 24mA, -25% to 125% | Â±(0.075% 1 digit) or Â± 3 digits, whichever is greater |

Voltage | 0 to 10V | Â±(0.075% 1 digit) or Â± 3 digits, whichever is greater |

Type J | -58 to 1830Â°F (-50 to 1000Â°C) | Â±(0.15% 1.8Â°F or 1Â°C) |

Type K | -58 to 2498Â°F (-50 to 1370Â°C) | Â±(0.15% 1.8Â°F or 1Â°C) |

Type T | -184 to 752Â°F (-120 to 400Â°C) | Â±(0.15% 1.8Â°F or 1Â°C) |

Loop Power | 24V | Â±(10ÂµV Â± 1 digit) |

Max. Load | 1000Î© @ 24mA | |

DC Measure | ||

Current | 0 to 50mA, -25% to 230% | Â±(0.075% 1 digit) or Â± 3 digits, whichever is greater |

Voltage | 0 to 20V | Â±(0.075% 1 digit) or Â± 3 digits, whichever is greater |

Type J | -58 to 1830Â°F (-50 to 1000Â°C) | Â±(0.15% 1.8Â°F or 1Â°C) |

Type K | -58 to 2498Â°F (-50 to 1370Â°C) | Â±(0.15% 1.8Â°F or 1Â°C) |

Type T | -184 to 752Â°F (-120 to 400Â°C) | Â±(0.15% 1.8Â°F or 1Â°C) |

Dimensions/Weight | 3.8 x 4.7 x 2.0″ (96 x 118 x 45mm)/12oz (340g) |